What Is Vacuum Degree Measurement Tester?
Arc quenching is carried out in a vacuum environment. The opening and closing of the current carrying contacts takes place in the vacuum chamber of the breaker, which is called the vacuum breaker. A vacuum used as the arc extinguishing medium in a circuit breaker is known as a vacuum circuit breaker because the vacuum provides high insulation strength due to its superior arc extinguishing properties.
Vacuum Degree Measurement Tester is a common method used to evaluate whether the vacuum degree of vacuum tube is broken. This method is an arc quenching chamber that can only assess the serious deterioration of the vacuum degree. When the vacuum degree drops to 10-2 — 10-1 Pa, the arc extinguishing chamber has already failed, although the breakdown voltage does not decrease.
MTV-4000 vacuum switch vacuum degree tester adopts a new excitation coil, and uses the magnetron discharge method to test the vacuum degree of the arc extinguishing chamber without disassembling the arc extinguishing chamber. At the same time, the microcomputer is used for synchronous control and data acquisition and processing, so that the field test sensitivity of the vacuum degree of the arc extinguishing chamber reaches 10-5 Pa. The most prominent feature of this instrument is the use of a new excitation coil and data processing method to achieve the measurement of vacuum without disassembly. This instrument has the advantages of convenient use, simple operation, no disassembly measurement and high test accuracy. It is a practical testing instrument widely used in power, steel, petrochemical, textile, coal, railway and other departments that use vacuum switches.
|Various types of vacuum switch tubes
|The new excitation coil is used for the measurement of the vacuum tube without
|Scope of Application
|This instrument is a multi-purpose machine, which can measure various types of vacuum
|10-5~ 10-1 Pa
|10-5~10-4 Pa, 10%
10-3~10-2 Pa, 10%
10-2~10-1 Pa, 10%
|Magnetic Field Voltage
|High Voltage of Pulse Electric Field
|The Opening Distance of the Switch Tube When Testing the Vacuum Degree
|Normal use opening distance